Double Beam Spectrophotometer U-2900/2910
Spectrophotometer mounting a large-size color LCD, which is simpler to use and higher in dependability.
High resolution satisfying European Pharmacopoeia (spectral bandpass: 1.5 nm).
Trace-amount measurement in biotechnological field, etc. can be carried out using optional 50, 25 and 5 micro cells.
Stable optics due to double beam
In double beam design, the energy of the light source is divided into two with a half mirror so that one passes through the reference side, and the other through the sample side, which is unavailable with the single beam design.
Since the reference-side energy is also incident on a detector, photometry is carried out on the basis of this signal. Therefore, an energy change in the light source can be compensated to ensure stable measurement for a long time.
Incorporation of stigmatic concave diffraction grating
The optics of this instrument adopts the Seya-Namioka monochromator widespread as a representative concavediffraction grating monochromator.
Because a concave diffraction grating has both beam condensing and dispersing functions, an optical system canbe configured with fewer mirrors.
In a spectrophotometer, use of fewer mirrors signifies a shorter optical path,thus giving rise to an aberration-free bright optics.
For elimination of the aberrations which were essentially unavoidable in the past, a stigmatic concave diffractiongrating has been developed by applying Hitachi’s original technology. As a result, a higher resolution has been realized.
A higher resolution has been achieved by eliminating coma from the Seya- Namioka monochromator which is the most popular concave diffraction grating monochromator. Its grooving is supported by the only ruling engine in Japan. The diffraction gratings of Model U-2900/2910 have also been made with this machine.
Spectral bandpass 1.5 nm satisfying European Pharmacopoeia
European Pharmacopoeia requires a ratio of 1.5 or larger when measuring 0.02% (V/V) solutions of toluene in hexane. In the spectrum at right, a satisfactory value of 1.6 can be confirmed.
(Sample: 0.02% solution of toluene in n-hexane)
Personal computer-based instrument control and data processing
U-2900 can be controlled by a PC equipped with optional UV Solutions Program. UV Solutions Program provides not only basic functions such as photometry quantitative analysis and wavelength scan but also the capability of data handling as described below.
(U-2910 is the model exclusively for operation with a PC. A PC equipped with Windows® is required for UV Solutions Program.)
Simple to use
Buttons are laid out so as to follow the operating procedure.
Hence, operations can be performed smoothly.
A rich variety of data processing functions
Spectrum can be expanded, contracted, smoothed, differentiated and integrated, and fundamental arithmetic calculations are applicable between spectra.
Powerful support for report generation
Measured data can be sent to Microsoft® Excel®. One click of the Report button suffices for data transmission.
A spectrum can be pasted to another application program.
Before printing, the contents can be checked by the Print Preview command.
Measured data text convertible
Besides conversion into the ASCII text or JCAMP format, spectra can be saved in Metafile format.
|Optical system||Double beam|
|Wavelength range||190 to 1,100 nm|
|Spectral bandpass||1.5 nm|
|Display||U-2900: color LCD with backlight (26.4 cm)|
|Printer I/F||U-2900: Centronics interface|
|Serial I/F||RS-232C (exclusive for UV Solutions program)|
|Size (main unit)||U-2900: 500 (W) × 605 (D) × 283 (H) mm (with LCD lowered)|
|U-2910: 500 (W) × 605 (D) × 241 (H) mm (without PC and printer)|
|Weight (main unit)||U-2900: 31 kg, U-2910: 29 kg|
|Power supply||100, 115, 220, 230 or 240 V, 50/60 Hz|
|Power consumption||300 VA|
|Option package program
|Capable of performing color calculation and color difference calculation.|
|Nucleic acid measurement program
|Usable for checking extraction and refinement of nucleic acids such as DNA/RNA essential for genetic research. Data (photometric values at 230, 260 and 280 nm), and calculation results (260/280, nucleic acid concentration, protein concentration, molar concentration) can be displayed collectively on the screen.|
|The burden of optical instrument performance check can be alleviated.
Item-dependent measurement mode and automatic measurement mode (when no sample is necessary) have also been prepared.
In addition, a judgment tolerance range is settable so as to meet each analytical purpose.
|Report generator program
|Developed to enable freely customizing the report output format of measurement results. Each character size
and position of report items and comment are settable and besides, mathematical calculations conventionally
dependent on manual method can be executed by spreadsheet program functions.
|UV Navigation program
|Provides support for automation of measurement, data processing and printout.
Measuring procedure can be readily programmed using a graphical editor for implementation of spectrophotometer control, measurement result indication and judgment, etc. Because analytical procedure can be automated, it is possible to save labor and form working patterns. This is effective for routine work such as quality control.