Skip to main content
In July 2019, an amendment to the RoHS directive will add 4 phthalates to the list of restricted substances. These 4 chemicals are widely used for insulated coating materials, electrical insulation tapes and packing films.
The HM1000A is a dedicated instrument designed for fast, easy, on-site screening of phthalates. In addition to easy operation, the HM1000A features high-throughput testing that achieves automated analysis of 50 samples in approximately 8 hours.
Vaporized compounds are ionized and directly analyzed by a mass spectrometer enabling high-speed measurement.(Separation column is not required)
The accurate and smooth performance of the auto sampler (standard equipment) allows continuous automatic analysis of up to 50 samples. During the analysis, operators are free to engage in other tasks, resulting in a streamlined operation and reducing workloads.
Place the sample in the sample pan and position it on the sample tray. Next, by following simple software steps, measurements will be started by the auto sampler. The auto sampler diagram is displayed on the software and the progress of measurements and the results (PASS/FAIL judgment) can be verified easily.
Nitrogen gas is used as the carrier gas. Using nitrogen gas generator (option available in Japan or locally sourced), a power source is the only utility required for operation. Helium gas and liquid nitrogen are not needed, which helps keep operating costs low.
The m/z value obtained from analyzing the sample can be used to estimate the component that is contained in the sample.
Mass spectrum of sample containing 1000 mg/kg of Phthalates
A peak will appear at molecular weight +1 for each phthalate after protonation.
The concentration of the Phthalates contained in the sample can be estimated using the ion intensity profile and can be compared to the regulations acceptable limit.
Ion intensity profiles of DBP, BBP, DEHP
The measuring sample :Certified Reference Material NMIJ CRM 8152-a
Di (2-ethylhexyl) terephthalate (DOTP) is included in alternative plasticizers used as replacements for the Restricted Substances. DOTP has a similar structure to the Restricted Substances. The HM1000A is calibrated for these substances that have similar structures based on ion intensity correlations. This has improved the identification accuracy of HM1000A, resulting in accurate measurements.
The HM1000A achieves simultaneous SIM and SCAN measurements, thereby enabling measurements that reflect the periodicity in the mass spectrum. This improves the screening accuracy of samples that contain lubricants and other polymers.
|Configurations||Standard equipment||HM1000A (Main unit / Accessories)|
|General Specifications||Measuring object||Phthalate (DBP/DIBP, BBP, DEHP)
Brominated flame retardants (DBDE)
|Sample weight||0.2 mg ± 0.04 mg|
|Sample form||Solid, Powder|
|Auto sampler||Max.50 samples|
|Mass detection||Ion source||APCI (Atomospheric Pressure Chemical Ionization)|
|Mass spectrometer||Quadruple mass filter|
|Measurement mode||Scan mode
SIM (Selected Ion Monitoring) mode
(Max. 20 channels)
|Dimensions and weight||Main body||510 mm (W) ×615 mm (D) ×615 mm (H), 85 kg|
|Rotary pump||160 mm(W)×490 mm(D)×265 mm(H), 29 kg|
|Electric specifications||Input supply voltage||AC200 V to 240V±10% single phase|